In ‘Mechanical properties and in vitro degradation behavior of additively manufactured phosphate glass particles/fibers reinforced polyactide,’ authors Lizhe He, Jiahui Zhong, Chenkai Zhu, and Xiaoling Liu explore a new level of material for 3D printing with phosphate glass/polylactide (PG/PLA) composites for use in medical applications such as fabrication of customized bone fixation plates for repairing fractures.
While bone regeneration is an area of great interest in 3D printing and additive manufacturing, so is the more common element of healing breaks, as researchers continue to look for better ways to improve the process—often accompanied by a range of bone fixation plates, screws, pins, and rods. Materials are key, along with integrity in design. Implants must be biocompatible, but the process is seamless when they are biodegradable too, thus eliminating the need for surgery.
The materials were tested for suitable mechanical properties as well as in vitro degradation behavior after creating models designed with PTC Creo Parametric, which were then imported into Simplify3D and the PG/PLA composites were 3D printed on an Ultimaker 2+. With the ability to fabricate complex geometries, the researchers could also control the level of porosity for bioprinting and tissue engineering purposes.
“Comparisons were made with PLA, and PLA reinforced with different loadings of PG particles (PGPs) as well as composites with reinforcements of different geometries [PGPs or milled phosphate glass fibers (PGFs)].”
The aim was to evaluate the AM composites as fracture fixation plates. A three-point bending test was performed, along with in vitro degradation for examining the strength and hydroscopy of the composites. There was a pH value check, along with dynamic mechanical analysis, and fiber length and laser particle size analysis. Both microscopy and statistical analysis were performed also.
In continuing to compare with PLA specimens, the authors noted the following:
- Improved flexural modulus
- Reduced flexural strength
- Reduced strain at break
- Intensified effects with increased PGP loading
“Embrittlement and strength reduction are associated with of stress concentration and low interfacial strength. It is likely here that the stress concentration effect was augmented by the incorporation of particulate with sharp corners. With increased filler loading, stress concentration sites also increased and led to more pronounced strength reduction and the same effect on strain at failure,” noted the authors.
Here, the average fiber length was 54 μm, and median and mode of fiber length were even lower. In comparison to authentic cortical bones, the PGF 10 composite was noted by the researchers to be ‘a close approximation,’ although flexural modulus was found to be considerably lower.
“Stiffness matching is recognized as the ‘gold standard’ for bone fixation implants, as fixation implants with such mechanical properties are strong and stiff enough for the load‐bearing activities without leading to ‘stress shielding.’ As such, it is probably necessary to consider the use of higher/longer fiber loading for this type of application,” stated the researchers.
Continuous PGF/PLA composites are more ‘suitable,’ according to the authors, in regard to load-bearing fixation—a feature connected with continuous fibers leading to stiffness. The flexural modulus of these materials, however, was reduced by ~80% after 28 days of degradation. The PGF 10 composites lost ~30% of initial flexural modulus after a degradation period of 56 days. The rapid flexural modulus could have been a result of the fiber ends being exposed in degradation media.
“Based on the consideration of both the initial mechanical properties and the facility to produce composites with desired geometries straightforwardly, the additive manufacturing of PG/PLA composites exhibits good potential in the making of patient‐specific fixation implants for bone that has low demand for load‐bearing, for example, zygoma, ankle, and maxilla,” concluded the researchers.
“These bones have been previously reported to be successfully restored using PLA‐based biodegradable fixation devices. Compared to PLA alone, it was demonstrated that the incorporation of PGF enhanced the flexural modulus of implants. It is also anticipated that the degradation of PGF releases magnesium, calcium, and phosphate to upregulate bone regrowth. Moreover, the FDM process allows fixation implants with customized geometries to be built directly and may remove the need for contouring of implants for anatomic fit during the operation.”
A serious interest in 3D printing today translates into a serious interest in materials—and most likely composites too, as they are able to add significant strength and improved properties to prototypes and parts, including that of polymers, bioprinting applications, and metals like titanium. What do you think of this news? Let us know your thoughts! Join the discussion of this and other 3D printing topics at 3DPrintBoard.com.[Source / Images: ‘Mechanical properties and in vitro degradation behavior of additively manufactured phosphate glass particles/fibers reinforced polyactide‘]
You May Also Like
3D Systems Expands in Denver to Address Demand for 3D Printed Products
As part of its planned expansion, 3D Systems announced it would add 50,000 square feet of facility space to its FDA-registered and ISO 13485-certified Denver, Colorado site. Along with new...
Satori Set to Launch Kickstarter Campaign for New Industrial Resin 3D Printer
London-based tech company Satori launched its first resin 3D printer, the compact yet professional ST1600, in October of 2020, and introduced a partnership program at the same time in order...
6K to Develop Battery Materials with $25M Investment
After just a few years in existence, 6K has made itself increasingly well-known in the 3D printing industry with its unique metal materials production technology. The startup suggests that its...
Virtual Review: the Formlabs Fuse 1 SLS 3D Printer
The desktop fused filament fabrication (FFF) movement quickly saw a proliferation of low-cost FFF systems, which was followed by a similar trend with desktop vat photopolymerization machines. The low-cost selective...
View our broad assortment of in house and third party products.