However, 3D printing, which has been used in the past to help treat CTEV, can help democratize the manufacture of these FAOs for more sustainable development, in addition to bringing the cost down. A new paper, titled “Open-Source Three-Dimensional Printable Infant Clubfoot Brace,” was recently published in the Journal of Prosthetics and Orthotics by a team of researchers, including open source advocate Dr. Joshua Pearce, from Michigan Technological University (MTU).
The researchers explain that inexpensive, open source RepRap 3D printers have helped expand access to the technology, while also keeping costs down.
“To investigate the potential of distributed manufacturing of an FAO for clubfoot patients in the developing world, this study makes a careful investigation of the use of RepRap 3-D printers to fabricate an open-source clubfoot brace,” the researchers wrote.
“The methodology includes first selecting among the various commercial 3-D printing materials based on material properties and cost, developing an open-source design using only open-source tools, and describing the open-source 3-D printer used and the settings to fabricate the clubfoot brace. Then, the brace’s features were examined to ensure that it met the required features for foot abduction orthoses designed for treating clubfoot. Finally, this brace was then examined by doing a cost and functionality comparison with existing designs.”
The team designed their 3D printable clubfoot brace using free, open source FreeCAD software, in order to “enable customization.” The design was optimized for easy printing, with minimal overhangs, and was “created with the intent of being able to meet criteria collected from the literature.” The most important features in the brace are:
- the ability to maintain foot positioning in order to prevent relapses
- the ability to adjust the angle of abduction
- the ability to dorsiflex the feet
Additionally, by making sure the width of the FAO between the feet is adjustable, the brace will need to be replaced less, thus making it more cost-effective.
“The ability of the foot pads to move independently and attach and detach from the abduction bar increases comfort and encourages proper usage,” the researchers explained.
Due to its strength and low cost, the researchers chose to 3D print the brace, and its 13 assembly components, out of biodegradable PLA. Cura software was used to slice the CAD models, and the brace was printed, at a speed of 40 mm/second, on a MOST delta RepRap 3D printer with an infill density of 80%, 0.2 mm layer height, and a 1 mm shell thickness.
Then the team ran a cost analysis on the three main variables – labor, filament, and electricity – that would drive the cost of manufacturing the brace locally in Kenya, where CTEV is “especially prevalent.” Labor costs in the country varied from $0 for volunteers, or employees who simply monitor the 3D printer and assemble the brace, up to the government-dictated wage of $1.30/hour for a Kenyan machinist in an urban area. Three commercial filament sources are available, while electricity costs run from $0/kWh for running a solar-powered 3D printer to $0.2029/kWh – the highest standardized rate for electricity in Kenya based on data from January of 2017.
The researchers found that the total material cost of the clubfoot brace was $11.08. However, it should be noted that the variable with the most potential to effect the overall cost is the labor, though electricity and filament costs can also vary.
The base of the FAO consists of top and bottom sliders held together by two carriage bolts, a closing bracket, and two nuts. By loosening and tightening the knobs, the feet width of the brace can be adjusted for a growing child.
“Due to the simplicity of the assembly, the infant’s shoes can be first attached to the footpad via Velcro straps before attaching the footpad to the bar,” the researchers explained. “The shoes employed should be “low rise” at the heel to allow observation that the heel remains against the foot pad, which is lost if a “high top” shoe is employed.”
This FAO offers more functionality at lower cost, and also has a lower environmental impact, as it only requires a partial infill and no shipping is required when it’s 3D printed locally. The team made the STL files and FreeCAD models for its infant clubfoot brace available for free at the Open Science Framework under a GNU General Public License 3.0.
The researchers concluded, “The results show that the open-source clubfoot brace matches or surpasses the physical features and mechanical degrees of freedom of all commercial- and non-profit-developed brace designs while substantially reducing the costs of the braces to hospitals and families.”
You May Also Like
Cartilage Tissue Engineering via Characterization and Application of Carboxymethyl Chitosan-Based Bioink
International researchers continue the trend in exploring natural biomaterials for bioprinting, detailing their findings in the recently published ‘Characterization and Application of Carboxymethyl Chitosan-Based Bioink in Cartilage Tissue Engineering.’ Examining...
Review: Current Efforts in 3D Printed Pharmaceuticals and Drug-Delivery Systems
International researchers continue to explore the potential of patient-specific treatment via 3D printing, releasing their findings regarding pharmaceuticals in the recently published ‘3D Printing of Pharmaceuticals and Drug Delivery Devices.’...
Regemat3D Launches its New Bioreactors for Maturing Tissues
One of Spain’s leading biotech companies, Regemat3D, has been developing custom biofabrication systems and regenerative medicine solutions since 2011 to fulfill unique research requirements and offer customized solutions for patients’...
Nanyang Technological University: Thesis Validates Use of Bessel Beams in Laser-Based 3D Printing
Andy Wen Loong Liew has submitted a thesis, ‘Laser-based 3D printing using bessel beams for tissue engineering applications’ to Nanyang Technological University. Exploring a new technique for bioprinting, Liew studies...
View our broad assortment of in house and third party products.