Microneedle arrays, or MNAs, are devices made up of micron-sized needles that make it possible to transfer a signal or compound across an outer layer of tissue, like skin. Because of how tiny they are, MNAs are a fairly painless and minimally invasive device, and have applications in chemical biosensing, drug delivery, and electrical stimulation, among others. Many materials can be used to fabricate microneedles, and based on the amount of applications they can be used for, it’s important to find a more cost-effective way of making them, which is what researchers Kevin J. Krieger, Nicky Bertollo, Manita Dangol, John T. Sheridan, Madeleine M. Lowery, and Eoin D. O’Cearbhaill from University College Dublin set out to do in their paper, “Simple and customizable method for fabrication of high-aspect ratio microneedle molds using low-cost 3D printing.”
The abstract states, “We present a simple and customizable microneedle mold fabrication technique using a low-cost desktop SLA 3D printer. As opposed to conventional microneedle fabrication methods, this technique neither requires complex and expensive manufacturing facilities nor expertise in microfabrication. While most low-cost 3D-printed microneedles to date display low aspect ratios and poor tip sharpness, we show that by introducing a two-step “Print & Fill” mold fabrication method, it is possible to obtain high-aspect ratio sharp needles that are capable of penetrating tissue. Studying first the effect of varying design input parameters and print settings, it is shown that printed needles are always shorter than specified. With decreasing input height, needles also begin displaying an increasingly greater than specified needle base diameter. Both factors contribute to low aspect ratio needles when attempting to print sub-millimeter height needles. By setting input height tall enough, it is possible to print needles with high-aspect ratios and tip radii of 20–40 µm. This tip sharpness is smaller than the specified printer resolution. Consequently, high-aspect ratio sharp needle arrays are printed in basins which are backfilled and cured in a second step, leaving sub-millimeter microneedles exposed resulting microneedle arrays which can be used as male masters. Silicone female master molds are then formed from the fabricated microneedle arrays. Using the molds, both carboxymethyl cellulose loaded with rhodamine B as well as polylactic acid microneedle arrays are produced and their quality examined. A skin insertion study is performed to demonstrate the functional capabilities of arrays made from the fabricated molds.”
Important design criteria when making microneedles includes limiting needle height to the sub-millimeter range and getting the correct aspect ratio and tip radius for the needle. While MNAs can be created for direct application, the researchers chose to use them as a master for fabricating microneedle molds, which can be made from a wider variety of materials.
There are many ways to fabricate microneedles for MNA mold masters, such as UV lithography, chemical wet etching, centrifugal lithography, electrical discharge machining, and micromilling, but as most laboratories likely don’t have the necessary equipment to carry these methods out, the researchers chose to use SLA 3D printing – specifically, the Form 2 3D printer from Formlabs, using Clear Resin.
“In this study, we present a novel customizable method for fabrication of microneedle masters in the research setting, which may be used to produce female molds which aims to overcome limitations of previous microneedles produced using low-cost desktop 3D printers which often suffer from low tip radius and/or low aspect ratios,” the researchers wrote. “To improve feature resolution, we developed a two-step “Print & Fill” technique which allows for the indirect fabrication of customizable microneedle masters for mold making using SLA 3D printing. Subsequently, microneedle masters and molded microneedle arrays are dimensionally characterized and functionally assessed for reproducibility.”
The researchers conducted a parametric study so they could find the optimal settings for fabricating the best needle geometry. In the study, they “quantified print success” by measuring needle height, angle, tip radius, and needle straightness. The needles were 3D printed at an aspect ratio of 4:1, and the team found that the smaller the layer height, the smoother the surface of the needle; acceptable needle qualities could be achieved with 25, 50, and 100 -µm, but 25 was the best.
“The first measured design parameter was the output needle height, which was seen to be lower than the set input height value. While the printed needles were shorter than specified in the virtual model, this appeared to be a predictable factor,” the researchers explained.
“A second design parameter measured was the angle θ. It is proportional to the aspect ratio for a needle with infinite tip sharpness. This value quantifies the angle between needle sides irrespective of the tip radius. The actual aspect ratio of a real needle will be lower with respect to that of a perfect needle (i.e., a needle of infinite tip sharpness) due to its inherently decreased height resulting from tip rounding. This parameter was relatively proportionate to the input value for needles of greater input height.”
The 3D printed needles’ tip radii was between 20 and 40 µm, which is sharp enough to pierce skin.
“The straightness of the needle was also quantified by extracting the needle outline from the images, converting it to a set of data points, and subsequent calculating of the coefficient of determination (R2) for the data points for each needle side. A straighter needle will have a R2 value closer to 1. Calculations found this value to lie in the range of 0.95–0.99 for all needles indicating good straightness of the print.”
The researchers used these optimized settings – 25 µm layer height, 2.5 mm input height, and 4:1 aspect ratio – as input for designing their needle array basins, which are made up of an array of 5 × 5 needles. The basins were filled with a photocurable resin in order to create four types of arrays with needles in the desired sub-millimeter range. Fabricated MNAs were used to create silicone microneedle molds, which were quality-tested by making MNAs out of two types of polymers.
The researchers used the molds to create PLA microneedles, and also conducted a successful porcine skin insertion study with the PLA MNAs.
The team’s two-part, inexpensive, and customizable Print and Fill method can definitely be used by other microneedle researchers in order to complete in-house fabrication of master molds and parametric optimization of MNAs.
Discuss this and other 3D printing topics at 3DPrintBoard.com or share your thoughts below.
Subscribe to Our Email Newsletter
Stay up-to-date on all the latest news from the 3D printing industry and receive information and offers from third party vendors.
You May Also Like
Desktop Metal: AM 2.0 Highlights from the Formnext Show Floor
Formnext, the leading international platform for Additive Manufacturing and industrial 3D Printing, returned in full swing to the halls of the Frankfurt convention center in Germany this November. With challenging...
Desktop Metal Receives $9M 3D Printer Order from German Car Maker
Original equipment manufacturer (OEM) Desktop Metal (NYSE: DM) announced that the company has received a $9 million order from a “large German car manufacturer.” Although it is not clear which...
3D Printing Financials: Markforged’s Supply Chain Issues Hinder FX20 Production
Supply chain disruptions continue to torment the manufacturing industry. In additive manufacturing, the challenging operating environment is harming production continuity. For Markforged (NYSE: MKFG), in particular, these production hurdles slowed...
The 19 Most Famous Angel Investors in 3D Printing
You may have the greatest idea in the world and just need that small investment of faith to launch it into a fully-fledged business. If a single individual, perhaps with...