Boluses are used often to protect the body from radiotherapy targeted at specific areas, and 3D printing can help streamline their production. Four Australian researchers published a paper, titled “Comparison of 3D printed nose bolus to traditional wax bolus for cost-effectiveness, volumetric accuracy and dosimetric effect,” about their work evaluating the dosimetric and volumetric, and cost, differences between a 3D printed nose bolus and one made with traditional wax.
“Three‐dimensional printing allows the conversion of digital models from data sets created via 3D ultrasound, computed tomography (CT) and magnetic resonance imaging into physical objects,” the team wrote. “In radiotherapy, 3D printed bolus is gaining momentum as the ideal patient bolus due to its durability, and superior patient surface contact over manually manufactured wax or gel bolus.”
Because of its contour changes, the nose is one of the most difficult areas to treat when it comes to skin cancer. But, to ensure that a full dose of radiotherapy is delivered to the patient’s skin surface, along with a homogenous dose distribution to the nose, it’s important that the bolus covers and conforms to the nose.
It’s a labor intensive process to make a nose bolus out of wax, as the technician’s skill level can affect the dosimetric and volumetric accuracy, and there can also be variations from the process itself, like imperfect skin contact or air cavity flaws. But a 3D printed bolus doesn’t have these issues.
“This study aimed to examine the volume, dimensional and dose differences between manually manufactured wax bolus and a 3D printed shell to the virtual bolus demonstrated in a radiotherapy plan for photon treatment to the nose, and to explore which type of bolus is more cost‐effective to produce,” the researchers explained.
The team enlisted 24 volunteers, and took plaster impressions of their noses using alginate and Plaster of Paris. Each positive plaster impression was CT scanned in 2 mm slices, and the scans were imported into the treatment planning system (TPS), contoured, and then “3D planned with virtual bolus and parallel opposed 6 MV lateral fields with collapsed cone algorithm.” Each virtual bolus (VB) plan was manufactured the traditional way, using wax, and then 3D printed out of PLA on an Ultimaker 2+.
A duplicate was fabricated for each 3D printed shell in order to check for conformity of fit and reproducibility. The 3D printing parameters included:
- 0.2 mm layer height
- 0.5 mm external perimeter thickness
- 0.8 mm top/bottom thickness
- 100% and 18% fill density
“To compare volume variations and dosimetry, each constructed bolus was CT scanned and a plan replicating the reference plan fields generated,” the researchers wrote.
Each bolus was contoured, and then assessed via CT scanning for any volume variations from the “reference plan’s VB contours.” Then, the plaster impression plans were copied to the CT image data set for the two types of bolus, and the team recorded and compared the dose at four different points.
“Points A, B and C were placed as close as possible to the same position across plans, accepting differences in exact point position (real slice or between slices) and that the bolus was not perfectly straight on each scan,” the researchers explained. “Additionally, a pseudo‐PTV was created on the VB plan, copied to all other bolus data sets, and maximum and mean doses to this volume recorded.”
They also computed dose calculations for more dosimetric analysis – once on the 18% infill shells with water fill, and twice on the 100% shells, with water fill and then with wax fill. CT scans were performed, and the team calculated the difference in dose distribution for each defined point, and the maxium and mean doses to the pseudo-PTV for both the original reference plan and the boluses.
The researchers wrote, “Mean volume differences between the virtual bolus (VB) and wax, and the VB and 18% and 100% 3D shells were −3.05 ± 11.06 cm3, −1.03 ± 8.09 cm3 and 1.31 ± 2.63 cm3, respectively. While there was no significant difference for the point and mean doses between the 100% 3D shell filled with water and the VB plans (P> 0.05), the intraclass coefficients for these dose metrics for the 100% 3D shell filled with wax compared to VB doses (0.69–0.96) were higher than those for the 18% and 100% 3D shell filled with water and the wax (0.48–0.88).”
The manufacturing time, and material costs, of fabricating the boluses were also analyzed. The team focused on consumable costs for this, and did not include the 3D printer’s purchase cost, maintenance and future replacement of its parts, and the tools and water bath for the wax bolus. Three of the 48 3D printed boluses did not completely print, and time lost on these was also left out of the cost analysis.
You can see in the table below that average costs for materials and staff time were higher for the wax boluses than for the 3D printed ones. Average time to 3D print the 18% infill shells was 529 minutes, while it took 747 minutes to print the 100% infill shells.
“As the 18% shells were converted and completed prior to all the 100% shells, the total cost of staff time is higher for the 18% shell group due to staff being more familiar with the process of DICOM to .stl file conversion for the 100% shell group,” the researchers explained.
Overall, the 3D printed boluses for photon nose treatments were more accurate and cost-effective than their wax counterparts, and will likely result in a better experience for patients.
“When printed in 100% shell infill density, and filled with water, 3D bolus shells accurately replicate the reference plan dosimetry. Three‐dimensional printed bolus shells produce superior bolus geometry, are more cost‐effective to produce than traditional wax bolus and do not require the same technical skills for production as traditional wax bolus,” the researchers concluded.
Discuss this and other 3D printing topics at 3DPrintBoard.com or share your thoughts below.
You May Also Like
GE Additive, Siemens and 10 More Join Sustainable 3D Printing Trade Group
Securing environmentally friendly manufacturing industries is ideal to reduce cost and waste, support cleaner and greener technologies, and lower the negative impacts of production on society and the environment. Additive...
Registration Rates Go Up This Week for Additive Manufacturing Strategies 2021
Additive Manufacturing Strategies 2021, the annual summit on business intelligence for the additive manufacturing industry held by SmarTech Analysis and 3DPrint.com, is coming to a computer screen near you February 9-12, 2021,...
3D Printing Webinar and Virtual Event Roundup, October 17, 2020
This week is packed full of interesting webinars and virtual events covering a wide variety of topics, from EBM technology and medical devices to software and design for additive manufacturing....
Early Bird Rates for Additive Manufacturing Strategies 2021 End October 21st
This winter, Additive Manufacturing Strategies, the annual business intelligence for AM summit held by SmarTech Analysis and 3DPrint.com, will be moving to an online format. AMS 2021, “The Business of 3D Printing,”...
View our broad assortment of in house and third party products.