Researchers from Saudi Arabia and Manchester, UK explore the creation of bone plates via electron beam melting (EBM) processes, outlining their findings in the paper ‘Topology optimised metallic bone plates produced by electron beam melting: a mechanical and biological study.’
Authors Abdulsalam Abdulaziz Al-Tamimi, Boyang Huang, Cian Vyas, Miguel Hernandez, Chris Peach, and Paulo Bartolo remind us that while metallic bone plates may be commonly implanted after fractures, there are some major disadvantages—and mainly due to implant/bone mismatch which could restrict the healing process. Using EBM, the research team was on a mission to create plates with decreased stiffness but similar mechanical properties.
While locking compression plates are often used to treat bone fractures, challenges may be encountered when it is time for such implants to be removed. In some cases, they may even cause a re-fracture or damage to nerves. There is always the option of leaving the implant, but the researchers point out that this scenario offers the risk for corrosion, the release of metal ions into the body, and even the possibility of patient reactions to materials which may be carcinogenic. To improve on these options, the researchers combined the following in their study:
- Additive manufacturing
- Computer modeling
During this study, the experimental plates were created using Materialise Magics software, and then exported into Build Assembler for slicing and uploading. The researchers also created support structures for the plates to add stability, as well as prevent thermal stress and warping. The EBM Arcam A3 was used for fabrication of the plates.
Great care was taken not only to prevent mechanical failure but also deformation. Stiffness values characterized the bones plates, with four, six, and eight holes. In each sample, the researchers noted stiffness reduction after an increase in volume reduction.
“A good approximation between numerical and experimental results was also obtained, showing that the computational tools considered in this research represent a viable approach to design and simulate their behavior before expensive fabrication and experimental steps,” stated the researchers. “Results also show that topology optimization is a viable tool to redesign bone plates, reducing their equivalent stiffness and consequently the stress shielding effect.”
The study was restricted in ‘considering the mechanical stability of the topology optimized bone plates,’ but the researchers are addressing that with analysis of the plastic regions.
“Most of the bone plates presented a higher yield stress with two plates (four-hole 75% volume reduction and six-hole 25% volume reduction plates) showing similarity with the commercial ones in terms of yield behavior. The resulting yield stresses for the two bone plates that have similar equivalent Young’s modulus to the cortical bone showed approximately half of the commercial bone plate yield. However, if we consider that the maximum force required to yield the two bone plates are 4191 N (six holes and 75% of volume reduction) and 5481 N (eight hole and 75% of volume reduction), this is still clinically acceptable as it corresponds to 6 to 8 times the bodyweight of a 70-kg patient,” stated the research.
The plates fabricated via EBM were not required to undergo any post-processing at all. This meant the elimination of:
- Residual stresses
- Surface finishing
Because of the resulting surface roughness, there was good cell attachment, leading the researchers to believe that a lack of polishing was positive overall.
“Cell viability studies also show a significant number of dead cells in the commercial plates after 14 days of cell seeding compared to the EBM plates,” stated the researchers. “This can be partially explained by the surface topography of the commercial bone plates. However, further investigation is required to elucidate the role that ion release and wettability have on cell interactions.”
In conclusion, the researchers pointed out that EBM parts lacking in post-processing treatment were also more biocompatible than commercial plates.
While 3D printing has lent its magic to nearly every industry today in some form, the medical field has been impressively impacted—and especially in the areas of surgical implants, devices, and models. Researchers have been able to create unbelievable new options for patients, from dissolvable magnesium implants to self-folding implants, and even antibacterial 3D printing filament for creating safe medical devices. Find out more about bone plates created through EBM processes here.
What do you think of this news? Let us know your thoughts! Join the discussion of this and other 3D printing topics at 3DPrintBoard.com.[Source / Images: Topology optimized metallic bone plates produced by electron beam melting: a mechanical and biological study]
Subscribe to Our Email Newsletter
Stay up-to-date on all the latest news from the 3D printing industry and receive information and offers from third party vendors.
You May Also Like
Dental 3D Market Grew to $4B in 2022
SmarTech Analysis, the leading 3D printing market research firm and the sibling firm of 3DPrint.com, has released the latest iteration of one of its flagship reports, 3D Printing in Dentistry...
UltiMaker CEO Weighs in on the Release of the Method XL 3D Printer
Today, UltiMaker announced the release of the Method XL 3D printer. The XL features a 100°C heated chamber and heated build plate optimized for ABS and carbon fiber (CF) -ABS....
3D Printing News Briefs, May 17, 2023: Stress-Resistant Alloy, 3D Printed Trophies, & More
In today’s 3D Printing News Briefs, we’re starting off with a little research, as materials scientists developed a 3D printing process that produces an extremely stress-resistant alloy. Moving on to...
3D Printing News Briefs, May 13, 2023: RAPID Roundup Part 2
For the second time this week, 3D Printing News Briefs is focused on news stories about RAPID + TCT! From new hires and 3D printer integrations to new 3D printers,...
Upload your 3D Models and get them printed quickly and efficiently.