In the recently published ‘Effects of Printing Parameters on the Fit of Implant-Supported 3D Printing Resin Prosthetics,” authors Gang-Seok Park, Seong-Kyun Kim, Seong-Joo Heo, Jai-Young Koak, and Deog-Gyu Seo delve into a niche realm in 3D printing: dental prosthetics. These types of implants can be challenging in many cases, but new technology brings enormous potential to the dental field.
The authors investigated 3D printing parameters on both fit and internal gap issues in dental prosthetics fabricated with resin, fabricating samples to examine fit. In dental manufacturing, additive manufacturing processes offer the advantages of speed in manufacturing, less waste in use of materials like resin, as well as being quieter and putting out less heat in the end. Digital light processing is commonly used in dentistry, with a digital mirror device (DMD).
Previous research has been performed regarding the fit of such dental implants, but none so far regarding fit in connection with parameters like orientation and thickness. There are numerous ways to measure gaps, both marginal and internal, to include:
- Direct view technique
- Cross-sectioning technique
- Impression replica technique
- Optical coherence tomography
- Micro-computed tomography
A model was 3D printed for two implants with customized abutments for the mandibular second premolar and second molar area.
The researchers used a D2-120 DLP 3D printer, with five build orientations: 0°, 30°, 45°, 60° and 90°, and two-layer thicknesses of 50 µm and 100 µm.
“In the 0° build orientation, supporting structure was connected perpendicular to the occlusal surface of the prosthesis,” stated the authors. “In the 90° build orientation, supporting structure was connected perpendicular to the lingual surface of the prosthesis.”
In all, 100 samples were printed, along with ten milling prosthetics. They were all examined under a microscope for defects, with no adjustments made.
“At the 50 µm layer thickness, the 60° build orientation was the smallest followed by the 45° build orientation and at the 100 µm layer thickness, 90° build orientation was significantly smaller than the other build orientations,” stated the authors. “At 45° and 60° build orientations, the 50 µm layer thickness group was significantly smaller than the 100 µm layer thickness group, and at 90° build orientation, the 100 µm layer thickness group was significantly smaller than the 50 µm layer thickness group.”
The researchers theorized that the large cervical/axial gap was caused by the sizeable cement space. In noting that they also found variances for marginal fit and internal gap, the researchers realized several issues could be to blame: support structure connections, errors due to no support, the appearance of layer output due to 3D printer changes.
“Considering the marginal fit and the internal gap together, build orientation of 45° and 60° would be recommended. The marginal fit of 3D printed prosthesis with the 100 µm layer thickness was like that of the 50 µm layer thickness,” concluded the researchers. “When the prosthesis is manufactured by 3D printing with favorable parameters, it is possible to obtain a comparable fit to the prosthesis manufactured by the milling. The marginal fit of 3D printing and milling prostheses was within the clinically acceptable limit for marginal fit of conventional cast restoration.”
While 3D printing is making huge impacts in the dental field, a variety of prosthetics are being fabricated today—and making a huge difference in the lives of so many individuals, from soft tissue maxillofacial prosthetics to ocular prosthetics and a wide range of developments and choices for children. What do you think of this news? Let us know your thoughts! Join the discussion of this and other 3D printing topics at 3DPrintBoard.com.[Source / Images: ‘Effects of Printing Parameters on the Fit of Implant-Supported 3D Printing Resin Prosthetics’]
You May Also Like
Delft University of Technology & Maaike Roozenburg 3D Print Chinese Porcelain
China is famous for its blue and white porcelain, delicately and artfully produced and painted. Crafted mainly in the southern Chinese city of Jingdezhen and purchased by travelers visiting the...
3D Printing News Briefs: July 11, 2019
We’ve got plenty of new products to talk about in today’s 3D Printing News Briefs, starting with materials from two chemical companies. WACKER announced new grades of of liquid and...
How do 3D Printed Dentures Stack up when Compared to Milled and Injection Molded Dentures?
In a new study, Korean medical researchers have been looking into the differences in quality and accuracy of several different modern ways to make dentures, with a focus on whether...
Additive Manufacturing Strategies Boston 2019 Speaker Roundup
January 29 to 31st Boston will host the Additive Manufacturing Strategies event which will be a chance for you to learn with and from your peers in medical and dental...
View our broad assortment of in house and third party products.